Portable Vacuum Suction Unit Machine ASA01
portable vacuum suction unit machine ASA 01 Description of portable vacuum suction unit machine : The HINOR ASA01 is portable vacuum suction unit machine ,professional medical suctioning device that produces a maximum vacuum of 560 mm Hg. It features an anti-bacteria filter design helps prevent...
The HINOR ASA01 is portable vacuum suction unit machine ,professional medical suctioning device that produces a maximum vacuum of 560 mm Hg. It features an anti-bacteria filter design helps prevent contamination, a high quality vacuum regulator and anti-vibration vacuum gauge setting, oil-less, permanent lubrication, and a collection bottle with overflow protection.
1. New design of anti-bacteria filter.
2. High quality vacuum regulator and anti-vibration vacuum gauge setting.
3. Our model is Oil-less,and permanent lubrication.
4. overflow protection collection bottle design.
5. Superior performance and high reliability for continuous operation.
6. Easy for carrying leaning.
7. OEM sevice available.
(L) 375x (W) 175x (H) 250 mm (14.8x6.8x9.8 in)
5.5 Kg (12.1 lb)
560+/- 3% mm Hg
AC 115V/60Hz 220V/60Hz
Bacteria Filter pore size
Collection Bottle Capacity
800 ml (cc)
41° ~ 95° F (5° ~ 40° C)
Operating Atmospheric Pressure
0.2~15.4 psi (70~106 kpa)
14° ~158° F (-10° ~70° C)
1. Vacuum Gauge 8. Patient Tubing Connector
2. Vacuum Adjust Knob 9. Connection Tubing
3. Bacteria Filter 10. Power Switch
4. Cord Caddy 11. Power Cord
5. Base 12. Fuse
6. 800cc Collection Bottle
7. Patient Tubing
Picture of portable vacuum suction unit machine:
How to operate your heavy duty suction pump
a. Before use, check the specification label on the side of the suction unit to ensure that the voltage and current indicated on the unit correspond with the voltage and current available. Make sure the power switch located on the side of the unit is in the “OFF” position before connecting the unit to a power source.
b. Switch the power switch to “I” position (on), the unit start running.
|Off position||On position|
|“off” position||“on” position|
c. Adjust the vacuum level from 0 to 560 mm Hg by turning the vacuum adjust knob (clockwise to increase vacuum and counter clockwise to decrease vacuum). Referring to the vacuum gauge while
setting the desired level of vacuum. To accurately read the gauge, block the patient end of the hose or cap off the collection bottle and allow the gauge to reach a stable vacuum reading.
d. Place the patient tubing to the appropriate position of patient and start to suck sputum.
e. When the sputum reaches the safe full level of collection bottle (800 cc), the suctioning will stop sucking automatically (float shut-off).
f. Switch the power switch to “O” position (off), shut off the suction motor.
g. Remove the sputum from collection bottle after the motor stops turning completely. (Please refer to the section “CLEAN INSTRU- CTIONS” on page 7 for the directions of how to clean the collection bottle).
h. Turn on the unit and start to run the suction unit again.
Packing of portable vacuum suction unit machine: